13. September 2015 14:28 by Christian
in Test Methods
A burst arc occurs when a mechanical contact is open during the switching process. Burst sources:
• Circuit Breakers in electrical circuits
• High Voltage switchgear
• 110/230V Power Supply systems
• 24V Control Lines
A burst has a single pulse rise time/duration of 5 ns / 50 ns from a 50 Ohm source impedance.
Bursts of 15 ms duration with a repetition rate of 5 kHz (or 100 kHz) are applied every 300 ms.
Voltage test levels:
• Power ports: 0.5 KV, 1 KV, 2 KV, 4 kV
• Signal and Control ports: 0.25 KV, 0.5 KV, 1 KV, 2 kV
• Coupling method is used to transfer the transient to the DUT.
• Decoupling method is used to block the transient from entering the mains and damaging other equipment connected in the network.
• Power line coupling is done with direct CDNs (Coupling/Decoupling Networks).
• Signal line coupling is done with a CCC (Capacitive Coupling Clamp): two metal plates which sandwich the line under test (cable) to provide a distributed coupling capacitance.
Test waveform verification is mandatory prior to each test.
For equipment connected to power ports all lines are coupled simultaneously.